Make use of them. Select the PC from the Hardware tab. Rename each control to match the measurement that’s being configured, “Voltage Manual Range” for example. Rename the X and Y axis of the Histogram graph to correspond with the values being measured by clicking on axis label and replacing the text or by clicking on the axis label and changing the Name text at the top of the Configuration Pane on the right. Create a constant for the type terminal on the top left side of the Get System Directory Node. Expand the Build Array Node to add an extra terminal by dragging down with the mouse after seeing the expansion cursor appear on the bottom of the Node.
|Date Added:||11 February 2004|
|File Size:||50.42 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Back to Top 5. Connect the Signal terminal at the top left of the Histogram Node and connect it to the voltage wire. When I run any simple example program from the Keithley driver such as turn off display or measure resistancethe Keithley will “beep” and display the following “Error: Create a constant for the type terminal on the top left side of the Get System Directory Node. We’ll be focusing on the section containing the “Read Multiple Points ” Node.
I am pretty new to the world of LabView and I am very lost right now. You can do more than just check that it appears in MAX. I have no idea what to do or how to fix this.
From the Configuration pop-up look at the Installed drivers section to find and install the NI When writing an instrument control application for a serial instrument, you will communicate with a COM port rather than the serial instrument connected to the COM port.
How do I get setup properly? Add an input for the number of bins to create by right clicking the number of bins 10 terminal on the left hand side of the Histogram Node and selecting Create control. Before running the VI type in the desired file name.
Keithley 648X Series Meter
Reposition controls on the Diagram for Configure Output and Enable Output nodes to make room for extra controls. Also, how do I go about verifying that the connection is there and correct? View how to create a Format Specifier by selecting the Write Delimited Spreadsheet Node and selecting Online manual in the Configuration pane on the right.
Then wire together the error out terminal of the Write Delimited Spreadsheet Node to the error in ksithley of the Close Node. This combination of features allows you to use NI system SMUs in a wide range of applications; from materials research and parametric test to high volume production test of RF and mixed signal ICs.
Click the Keithley 24XX project to create a new copy. Open the Panel by selecting Panel in the View Selector.
Keithley SourceMeter – Instrument Driver – National Instruments
Sign up or log in Sign up using Google. Expand the Build Array Node to include multiple elements. Sometimes instrument drivers default to certain settings you may not desire and may not even know about until you start digging. Serial instruments are not visible in MAX or keithleu Hardware tab.
Use the following procedure to add manual inputs for Voltage, Current, and Resistance ranges Create an “Enable Auto Range” boolean keithhley by right clicking the “Enable Auto Range” boolean constant and selecting Change to control.
Back to Top 2.
Keithley X Series Meter – IEEE (GPIB), Serial Driver for LabVIEW – National Instruments
Any help would be very much appreciated because I have been trying to solve this for a week, with absolutely no luck. We want users to be able to choose to have the Auto Zero enabled, disabled, or run immediately so we will create a Control at the Autozero terminal.
NI TestStand labvirw at the top of a properly architected test system. I am currently studying how to control the voltage and the current of a source meter which is Keithley using Labview software. Do the same for actual maximum and actual minimum terminals on the right hand side of the Histogram Node. Joe Friedrichsen 1, 12 When I use the “search for instruments” function under tools and instrument driversLabView does not detect the Keithley.